Calibration of cyclic force with inertial force correction to a fatigue testing machine
AbstractThe accuracy between a dynamic force and a static force applied on a specimen by a fatigue machine is usually not the same. By establishing physical vibration models of fatigue machines, it is concluded that the error of a cyclic force is mainly caused by the inertial force of the vibration mass between the machine sensor and the specimen. After the inertial force is exactly corrected, the force displayed on the machine would be consistent with the real force on the specimen. A standard dynamic force calibration sensor (DFCS) with an inertial force correction method has been used to do calibration of fatigue testing machines in this paper. Compared with the replica test-piece method, the two calibration results are close to each other.
Authors who publish with this journal agree to the following terms:
- Authors retain copyright and grant the journal right of first publication with the work simultaneously licensed under a Creative Commons Attribution License that allows others to share the work with an acknowledgement of the work's authorship and initial publication in this journal.
- Authors are able to enter into separate, additional contractual arrangements for the non-exclusive distribution of the journal's published version of the work (e.g., post it to an institutional repository or publish it in a book), with an acknowledgement of its initial publication in this journal.
- Authors are permitted and encouraged to post their work online (e.g., in institutional repositories or on their website) prior to and during the submission process, as it can lead to productive exchanges, as well as earlier and greater citation of published work (See The Effect of Open Access).